- Product Introduction
- Operation Method
- Video demonstration
- Operation software
(1) Smooth, beautiful and personalized appearance design.
(2) Automatic measurement, acquisition and processing of data.
(3) Storage of up to 500 test results for easy query and printing.
(4) User-friendly operation software, customized test report.
(5) With connection to hospital information system.
① Warm up after power-on
Install the instrument in a suitable working environment and connect to the power. Turn on the power switch of the instrument. When it reaches a stable state after warming up for 30 minutes, it can be normally used.
a. Insert the attached standard background card into the measuring slot of the instrument to calibrate the background for 1000 seconds without inserting a card. The calibration result shall be BGC ≤ BGM. Insert the attached standard background card into the measuring slot for sample measurement (set the parameter MOD_CON to 0), and the measurement result C should be negative.
b. Insert the attached standard efficiency card into the measuring slot for sample measurement (set the parameter MOD_CON to 0), and the measurement result C should be within ±30% of labeled value.
When the conditions of a and b are met simultaneously, the instrument can be operated normally.
③ Sample collection:
a. The subject takes one 14C-Urea on an empty stomach or two hours after eating.
b. Sit for 15-25 minutes.
c. Unpack the collection card and connect the mouthpiece to the card body. The subject blows into the card through mouthpiece with appropriate force. The subject may exchange breath during blowing, but DO NOT draw in breath! Stop blowing when the card indicator turns from orange to yellow (about 1-3 minutes) or after 3 minutes even though color is not completely changed.
d. Fill in the information of subject on the collection card after sample breath is collected.
④ Sample testing:
Insert the sample card into measuring slot, and test will start automatically. The test result will be displayed on the screen when test completes.
⑤ Result printing:
Test report can be printed either from external printer or built-in micro printer.