- Product Introduction
- Operation Method
- Video demonstration
- Operation software
(1) New structural design with beautiful appearance and light weight.
(2) Adopting newly designed circuit and elements with reliable performance and high efficiency.
(3) Fully-upgraded operation interface, intuitive and simple.
(4) Automatic design for button-free measurement and easy operation.
(5) Automatic fault diagnosis, background count reduction and correction.
(6) Automatic presentation and printing of test results.
(7) Barcode scanning, connectable to Hospital/Laboratory Information System (HIS/LIS)
① Warm up after power-on
Install the instrument in a suitable working environment and connect to the power. Turn on the power switch of the instrument. When it reaches a stable state after warming up for 30 minutes, it can be normally used.
a. Automatic background calibration: Power on the instrument and the instrument will perform automatic background calibration for about 45 minutes. During this process, no operation can be performed on the instrument. The instrument will automatically detect and adapt to the environmental background radiation.
b. Manual background calibration: Press the “Test” button within 10 seconds after turning on the power of the instrument. The instrument will automatically adapt to the environmental background within 41 minutes and 40 seconds and it can be used after self-adaptation. The background should be less than 250.
c. Insert the attached standard efficiency card into the measuring slot for sample measurement, and the measurement result C should be within ±30% of labeled value.
When the conditions of a and c or b and c are met simultaneously, the instrument can be operated normally.
③ Sample collection:
a. The subject takes one 14C-Urea on an empty stomach or two hours after eating.
b. Sit for 15-25 minutes.
c. Unpack the collection card and connect the mouthpiece to the card body. The subject blows into the card through mouthpiece with appropriate force. The subject may exchange breath during blowing, but DO NOT draw in breath! Stop blowing when the card indicator turns from orange to yellow (about 1-3 minutes) or after 3 minutes even though color is not completely changed.
d. Fill in the information of subject on the collection card after sample breath is collected.
④ Sample testing:
Insert the sample card into measuring slot, and test will start automatically. The test result will be displayed on the screen when test completes.
⑤ Result printing:
Test report can be printed either from external printer or built-in micro printer.